How does Moku:Lab's Laser Lock Box apply gain to the input?
If you still have questions or prefer to get help directly from an agent, please submit a request.
We’ll get back to you as soon as possible.
Moku:Go Arbitrary Waveform Generator Moku:Go Data Logger Moku:Go Frequency Response Analyzer Moku:Go Logic Analyzer & Pattern Generator Moku:Go Oscilloscope & Voltmeter Moku:Go PID Controller Moku:Go Spectrum Analyzer Moku:Go Waveform Generator Moku:Go Power Supplies Moku:Go Digital Filter Box Moku:Go FIR Filter Builder Moku:Go Lock-in Amplifier
- Python API examples
The +24 dB and +48 dB input gain on Moku:Lab is implemented purely digitally. It is designed to reduce quantization error when the FPGA performs the calculation. We recommend using the maximum possible input gain that does not saturate the input signal. Please note the built-in probe points have a fixed bit depth. The quantization error may be exaggerated in the built-in oscilloscope. The underlying calculations may be performed at higher precision.