Waveform Generator sweep mode settings and trigger options
In Triggered Sweep Mode, when a trigger is detected, waveform generation will begin at the Start frequency and sweep to the End frequency over a given duration. The Start frequency is set in the output waveform, the Stop frequency and duration are set in Modulation settings.
The source of trigger events can be selected from the following options:
- Input 1: Trigger from the waveform characteristics of Input Channel 1.
- Input 2: Trigger from the waveform characteristics of Input Channel 2.
- Output: Trigger from the signal generated in the other Output Channel.
- Internal: Trigger from an internally generated signal.
- External: Trigger from the TTL-compatible EXT input on the back panel.